Receiving system error 1018
Receiving System error 1018 to call service. Unplugged and waited 5-10 minutes, machine still doesn’t work.
@corkie The following message is displayed on the LCD: "SYSTEM ERR 1018 Call service!"
The error is a Flow sensor fault. The solution is to Replace flow sensor. You can download the service manual from this page ResMed VPAP Auto 25 Repair to work on your CPAP. Ultimately you do want somebody (preferably a BioMed person) look it over after you do your repair.